Panorama Nova RP Full Face Masks w/Plug-In Connector Instructions for Use 1 March 26 2009 O2 TNG, MRT Trainee POM Whats What 1. Face piece Connector 2. Clip 3. Inner Mask 4. Visor 5. Clamping Frame 6. Front Strap 7. Head Straps
8. Soft body 9. Face blank 10. Button 2 March 26 2009 O2 TNG, MRT Trainee POM Use For masks with visor wiper: Soak the wiper with the de-misting agent. It must not be actuated when dry. Hold the full face mask ready: Undo the straps as far as possible. Put on the PSS BG 4
Closed-Circuit Breathing Apparatus, and press the plug-in connector until it engages 3 March 26 2009 O2 TNG, MRT Trainee POM Use Put on the mask: Unbutton the strap and place chin in recess. Fit straps round the head and pull them back until the front strap rests smoothly on the face.
Position the Mask 1. Tighten the neck straps uniformly 2. Tighten the temple straps uniformly 3. Tighten the front strap if head is fairly small. 4 March 26 2009 O2 TNG, MRT Trainee POM Check for Leaks Tightly compress both breathing hoses and breathe in until vacuum is produced.
Briefly hold your breath: the vacuum must be maintained. If it is not, the straps must be tightened further. The leak test must be repeated twice. The mask may only be used if perfectly tight! 5 March 26 2009 O2 TNG, MRT Trainee POM Wiper If the visor mists over, it can be cleared by wiping. The wiper is activated via a lever on
the outside of the visor. The wiper must then be returned to its basic position in the middle of the visor. 6 March 26 2009 O2 TNG, MRT Trainee POM After use Disconnect the PSS BG 4 closed circuit breathing apparatus: press the button and simultaneously draw the plug-in connector out of the face piece connector
Remove mask: Slide your index finger under the ends of the neck straps, press the clamps forwards with your thumbs and evenly pull the mask over your head and off your face. 7 March 26 2009 O2 TNG, MRT Trainee POM Cleaning and Disinfecting Cleaning: The mask must be cleaned whenever it has been used. Position the wiper in the middle of the visor. The mask must be thoroughly rinsed in lukewarm
water containing a little all-purpose Sekusept Cleaner and a cloth. Rinse under running water. Disinfection: The mask must be disinfected whenever it has been used. Only use approved disinfectants. Rinse under running water. Drying: Leave mask to dry in drying cabinet or in air. 8 March 26 2009 O2 TNG, MRT Trainee POM
Faye. Glenn . Abdellah. Opieka nad pacjentem w teorii pielęgniarstwa. NAUKA I WYKSZTAŁCENIE. 1919 r. 13 marca - urodzona w Nowym Jorku. ... Abdellah, FG, i Levin, E. (1957) Developing a measure of patient and personnel satisfaction with nursing care....
We all know the Network+ Certification objectives . Perspectives on Teaching Network+. This exam will certify that the successful candidate has the knowledge and skills required to troubleshoot, configure and manage common network wireless and wired devices.
Engineering & Computer Science Departments Instructions: Use the previous page as a template to create a 11"x17" page data sheet and name the file the same as project title. Create a project report in pdf format that including a description,...
Times New Roman Arial Wingdings ABB Logo ABB-A Segment-Clear.pot Photo Editor 3.0 Photo Body-in-White Standard Production Systems Body-in-White framing system Modulflex Standard Welding Cells Hemmers Relative Positionning: Flexplace
CPI more useful for international comparisons Recent UK Inflation What does the above diagram tell us? Real and Nominal Measurements We typically use money values to measure economic output, using e.g. the prices of goods We must however be aware...
How do I sign onto the Project? All Luton and Bedfordshire community pharmacies will automatically be opted in to the TCAM project.. If you wish to opt out of . TCAM . you must complete . an opt . out...